34 | P a g e © Adam Equipment Company 2010
12.3.1 F1 -CALIBRATION
To enter the calibration parameter, press the [Z/T] key when “F1” is displayed. The
scale will be calibrated using 2 masses of approxim ately 1/3Maximum and
Maximum. If the scale has been calibrated once the values will be stored. If this is
the first time the scale is calibrated the user mus t enter the values for the
NOTE: It is necessary to set the decimal point loc ation and the capacity before
calibration is possible.
The display will instruct you to remove any weight from the scale, “UnloAd”. Press
The display will tell you to add the first weight to the scale: “Ld 1” “ 10 kg” If
necessary change the value shown to match the weigh t to be used. Press [CE] to
clear the old value and then enter the new value. All values entered are in whole
Add the weight shown, wait for stability then press the [Z/T] key.
The display will tell you to add the second weigh t to the scale: “Ld 2 “ “ 30 kg”
Either add the weight shown, wait for stability the n press the [Z/T] key or press the
[Print/M+/Esc] key to accept the single point calibration and ret urn to weighing.
The display will show “PASS” if the calibration is OK.
Remove the weight.
12.3.2 F2–DECIMAL POINT POSITION
To set the value for the decimal point location. The options are 0, 0.0, 0.00,
To enter this parameter, press the [Z/T] key when “F2 dEC” is shown.
The display will show the current setting.
Press the [Pcs/
] key to change the value. Select from 0, 0.0, 0.00, 0.000, 0.0000
Press [Z/T] to accept the displayed value.