© Adam Equipment Company 2015 41 3016612482 Rev1-Aug15
larger number corresponds to a larger stable zone. Selectable values: 1, 2, 5
or 10 (divisions).
Recommended value for normal use: 1
AV ZER\bAV ZER\b AV ZER\b
ON = Auto-zero function. Selectable values: 1, 2,
5, 10 or 15 (divisions).
OFF = Auto-zero function disabled.
Recommended value for normal use: ON, 5
COMMA Set separator indicator on the display to
be either a decimal point
DEC PT or a comma. Also applies to the serial interface for print output.
13.6 CALIBRATION SETUP
This menu allows the Supervisor to set the calibration parameters. Press [Select] when
“ CAL SET
AL SETCAL SET
” is displayed to select the calibration parameters. The options for each
parameter can be scrolled through by using the [Up] or [Down] key and pressing [Select]
to confirm choices.
NO = Operator calibration is disabled.
ES = Operator calibration is enabled.
CAL REPCAL REP CAL REP
ON = Prints out Calibration report after successfu
OFF = Disabled.
TIM CALTIM CAL
ON = Enabled. Select time from 1 to 24 hours.
FF = Disabled.
TEM CALTEM CAL TEM CAL
ON = Enabled. Select the temperature variation from 0.2 to 4°C which when
detected will trigger automatic calibration.
OFF = Disabled.
INT CALINT CAL INT CAL
YES = Internal calibration enabled (if fitted).
O = External calibration enabled.
INT MASINT MAS INT MAS
CAL MAS = Displays the set value of the internal cal
ibration mass (if fitted) in
grams. If after verification against an external mass it is determined that the
value of the internal mass needs adjustment, e.g. due to wear, accumulation
of dirt, etc., then this value can be adjusted by +/- 100 mg. This should only
be considered by expert users if the external reference weight is definitively
accurate and an incorrect weight reading is being given after internal
calibration. Adjustment will restore the internal calibration to the correct level
Press [Mode] to advance to setting of the next menu or [ 0/T ] t
o return to normal
To enable the security features in this balance it is necessary to set passcodes. There are
2 passcodes called Operator Passcode and Supervisor Passcode. The Operator
Passcode allows an authorised user to operate the basic weighing functions of the balance
but will not allow access to the Supervisor Menus if the Supervisor Passcode has been
Note: To change or disable a Passcode it is be necessary to enter the current passcode.
To setup passcodes:
Press [Select]. Use the [Up] and [Down] keys to cycle through options until “ PASSC\bd
isplayed. Press [Select] again to enter this section. Use [Up] and [Down] keys to select
Operator (“ PC \bPER
C \bPERPC \bPER
”) or Supervisor (“ PC SV
P C SVPC SV
PEPE PE”) option.
C O P Y